The sensor/probe consists of a non-linear thermistor configured with a precision resistor in a half-bridge circuit, as shown in the product’s manual:

- Model 107 Temperature Probe Instruction Manual
- Model 108 Temperature Probe Instruction Manual
- 108-LC Temperature Probe for MetData1 Instruction Manual
- Model 109 Temperature Probe Instruction Manual

To measure the sensor/probe, the measurement device has to provide a precision excitation voltage (Campbell Scientific data loggers use 2000 mV), measure the voltage across the precision resistor, determine the thermistor resistance (Ohm's law), and convert the resistance to temperature using the Steinhart-Hart equation.

The Steinhart-Hart equation is 1/*T* = *A* + *B*ln(*R) *+ *C*(ln(*R*))^{3} where:

*T*is the temperature in Kelvin*R*is the resistance at*T*in ohms*A, B,*and*C*are the Steinhart-Hart coefficients, which vary depending on the temperature range of interest, as well as the type and model of the thermistor

For the 107-L, 107-LC, 108-L, and 108-LC, the following are the coefficients for the Steinhart-Hart equation:

*A*= 8.271111E-4*B*= 2.088020E-4*C*= 8.059200E-8

For the 109-L, the following are the coefficients for the Steinhart-Hart equation:

*A*= 1.129241E-3*B*= 2.341077E-4*C*= 8.775468E-8

This was helpful

FAQs Home

- How can LoggerNet be upgraded to LoggerNet Admin?
- How much does a typical system cost?
- Can one license of LoggerNet Remote be installed on multiple computers in a single network?
- What is needed to connect a sensor that is compatible with wireless sensor interfaces to a CR800, CR1000, or CR3000 in a wireless network?
- What are the different cable termination options available for Campbell Scientific sensors?
- What type of field check can be done on a temperature sensor or a temperature and relative humidity sensor to determine if it is within the range of specified accuracy or if it requires calibration?
- What is a PRT temperature measurement?