Frequently Asked Questions About our Products and Solutions
Submit an enquiry
Submit an enquiry
(1972 total: showing 1-20)
Comment établir une connection avec un système d’acquisition de données à travers le port de série RS-232 (Pour le CR800, CR1000X et CR3000)
How to connect to a datalogger through an RS232 port? (For CR800, CR1000X and CR3000)
How quickly can measurements be attained using the ClimaVUE™50?
Does the wind gust measurement interval change for the ClimaVUE™50 based on the query interval?
With the ClimaVUE™50, over what period is wind gust measured?
Is the calibration for the CS240DM NIST traceable?
Are the traditional methods of using RTD to measure the back-of-module temperature not used with the CS240 and CS240DM because these methods may cause hot spots?
Can multiple instances of Campbell Scientific SURVEYOR be run on a PC at the same time?
How tight do the wing-nuts on the CM226 integrated mounting stand need to be tightened to secure the solar sensor?
Can the SG000 determine the exact distance to a lightning strike?
When selecting a recalibration service for a pyranometer, how do the services differ and what is the importance of ISO 9060?
What are some distinguishing features of the MetSENS compact weather sensors?
Can the CSAT3BH heaters be controlled manually?
What happens to the data when the CSAT3BH heaters are on?
How often should the CR1000 be recalibrated?
When should a data acquisition (DAQ) system be used instead of a PC-based system?
How can the age of a sealed rechargeable battery be determined?
If the CS135 reports a warning or alarm, when should a visit to the sensor be made?
Are the CS135 warnings and alarms configurable?
What does a CS135 alarm mean, or how is it defined in the message string?